"Really Really Cheap Digital Pocket Scale"
This Generic pocket scale has been sent to us by three different companies in the past. We see it on eBay and various websites for incredibly low prices, so we thought we should give it a review. The design is very basic, just two keys and a simple flip up cover. The scale is very small and easily fits in a front pocket.
This scale is another bare-bones model in the marketplace. It is sold by various companies under various brands but it's all the same basic scale. For purposes of this article we will call it the Digi-200 (that's what 2 of the samples say on them). The scale survived our drop tests but not very well. The magnet on the lid came off and became the first casualty, then the battery door splintered.
The performance of the Digi-200 was reasonably good. We calibrated the scale and then ran through our standard series of tests. All numbers were within +/- 0.2 gram as long as we placed the items in the center of the tray. The tray is very small which skewed our 4 corner test because the 4 corners were so close to each other (more on this later). Drift was minor and within tolerance (no drift within 40 seconds).
Our main complaint about this scale is it's tiny recessed weighing tray. We found it impossible to weigh a bag of grains because the bag would always touch the sides of the scale, distorting the results. The Digi-200 comes with a small items tray but again it's so small and awkward that we couldn't weigh anything other then small rings and gems. Anything larger is extremely difficult to weigh.
Recommendations: As noted above, we felt the tray was the Achilles heel of this scale.
FeedbackWe encourage you to provide feedback on this article and/or tell us your personal experience with a scale.
*Within reason - we stepped on the platform, dropped the scale from 4 ft, all without any damage except for cosmetic.
Important Disclaimer: Scales are sometimes difficult to compare. Our results are just a sample of tests done at random conditions. The actual operating conditions of your scale may vary from our test conditions and your results may differ from our results. We did our best to test the scales in a blind and unbiased manner. We receive advertising and other consideration from companies that support this site. We do our best to not allow this to affect the results of our tests and evaluations. However, we strongly recommend that you do your own comparison tests in your actual operating environment to determine which scale is best for your needs.
The terms PalmScale 7.0 and all are trademarks or trade names of their respective parties. Names are used for comparison purposes only and for the purposes of this article.